A Comparative Study Between Newly Developed Wave-form and Traditional Straight-form Splints
Kim, S; Jeon, W; Lee, J; Lee, CA; Choi, S; Jung, Y; Kim, G; Chung, WG
Taehan Ŭnggup Ŭihakhoe chi, 23(6):869-873, 2012
Taehan Ŭnggup Ŭihakhoe chi; Journal of the Korean Society of Emergency Medicine; 대한응급의학회지
Purpose: Widely used fiberglass splints are made from straight-form material. These prove difficult to mold at joints and form wrinkles, causing complications such as pain, pressure sores, and peripheral nerve palsy. We compared the usefulness of wave-form splints with straight-form splints and the level of satisfaction of these designs from care providers and wearers.
Methods: Eighty-nine (n=89) emergency physicians and orthopedic surgeons participated in this study. The subjects (acting as care providers and mock patients) used waveform and straight-form material to construct and wear short leg splints, long arm splints, and sugar tong splints. The clinicians were surveyed on their satisfaction as providers and wearers. All questions were measured on scores from 0 and 10(10=maximum score). After splints were removed, subjects were surveyed on the extent of splint wrinkling with scores from 1 to 3(3=maximum wrinkling).
Results: Provider satisfaction scores for wave-form splints versus straight-form splints in short leg splint applications were 7.76±1.30 vs 6.74±1.25 (p=0.000). Provider satisfaction scores for wave-form splints versus straight-form
splints in long arm splint applications were 7.73±1.33 and 6.73±1.59 (p=0.004), respectively. The subjects felt more comfortable wearing wave-form splints, compared to straight-form splints (7.79±1.49 vs. 6.79±1.58, respectively; p=0.004) and more satisfied (8.03±1.35 vs. 7.18±1.33, respectively; p=0.003). The frequencies of wrinkle occurrence in wave-form and straight-form splints were 29.7% and 42.2%, respectively (p=0.02).
Conclusion: Wave-form splints may be more practical for molding and wearing than traditional straight-form plints.
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.
Total Visit :3,087,098
Total Download :1,259,646
Today View :884
Ajou University Medical Information & Media Center 164 Worldcup-ro Yeongtong-gu Suwon 16499 Korea / TEL : 031-219-5312 / FAX : 031-219-5314 Copyright (c) Ajou University Medical Information & Media Center All Rights Reserved. AJOU Open Repository는 국립중앙도서관 OAK 보급사업으로 구축되었습니다.